PriceList for Fob Tumor Marker Rapid Test Kit - Diagnostic Kit for Progesterone (fluorescence immunochromatographic assay) – Baysen
PriceList for Fob Tumor Marker Rapid Test Kit - Diagnostic Kit for Progesterone (fluorescence immunochromatographic assay) – Baysen Detail:
Diagnostic Kit for Progesterone(fluorescence immunochromatographic assay)
For in vitro diagnostic use only
Please read this package insert carefully prior to use and strictly follow the instructions. Reliability of assay results cannot be guaranteed if there are any deviations from the instructions in this package insert.
Diagnostic Kit for Progesterone (fluorescence immunochromatographic assay) is a fluorescence immunochromatographic assay for the quantitative detection of Progesterone (PROG) in human serum or plasma, it is used for auxiliary diagnosis of progesterone abnormally associated diseases.All positive sample must be confirmed by other methodologies. This test is intended for healthcare professional use only.
Progesterone is an important hormone that plays an important role in the regulation of the menstrual cycle and is essential in maintaining pregnancy. The progesterone concentration in the serum increased rapidly after ovulation. It is a reliable indicator of natural ovulation or induction of ovulation.
PRINCIPLE OF THE PROCEDURE
The membrane of the test device is coated with the conjugate of BSA and PROG on the test region and goat anti rabbit IgG antibody on the control region. Marker pad are coated by fluorescence mark anti PROG antibody and rabbit IgG in advance. When testing sample, PROG in sample combine with fluorescence marked anti PROG antibody, and form immune mixture. Under the action of the immunochromatography, the complex flow in the direction of absorbent paper, when complex passed the test region, The free fluorescent marker will be combined with PROG on the membrane.The concentration of PROG is negative correlation for fluorescence signal, and the concentration of PROG in sample can be detected by fluorescence immunoassay assay.
REAGENTS AND MATERIALS SUPPLIED
25T package components：
Test card individually foil pouched with a desiccant 25T
Sample diluents 25T
Package insert 1
MATERIALS REQUIRED BUT NOT PROVIDED
Sample collection container,timer
SAMPLE COLLECTION AND STORAGE
1.The samples tested can be serum, heparin anticoagulant plasma or EDTA anticoagulant plasma.
2.According to standard techniques collect sample. Serum or plasma sample can be kept refrigerated at 2-8℃ for 7days and cryopreservation below -15°C for 6 months.
3.All sample avoid freeze-thaw cycles.
Please read the instrument operation manual and package insert before testing.
1.Lay aside all reagents and samples to room temperature.
2Open the Portable Immune Analyzer(WIZ-A101), enter the account password login according to the operation method of the instrument, and enter the detection interface.
3.Scan the dentification code to confirm the test item.
4.Take out the test card from the foil bag.
5.Insert the test card into the card slot, scan the QR code, and determine the test item.
6.Add 20μL serum or plasma sample to sample diluent, and mix well..
7.Add 80μL sample solution to sample well of the card.
8.Click the “standard test” button, after 10 minutes, the instrument will automatically detect the test card, it can read the results from the display screen of the instrument, and record/print the test results.
9.Refer to the instruction of Portable Immune Analyzer(WIZ-A101).
TEST RESULTS AND INTERPRETATION
follicular phase/ovulatory period
.The above data is the result of PROG reagent test, and it is suggested that each laboratory should establish a range of PROG detection values suitable for the population in this region. The above results are for reference only.
.The results of this method are only applicable to the reference ranges established in this method, and there is no direct comparability with other methods.
.Other factors can also cause errors in detection results, including technical reasons, operational errors and other sample factors.
STORAGE AND STABILITY
1.The kit is 18 months shelf-life from the date of manufacture. Store the unused kits at 2-30°C. DO NOT FREEZE. Do not use beyond the expiration date.
2.Do not open the sealed pouch until you are ready to perform a test, and the single-use test is suggested to be used under the required environment (temperature 2-35℃, humidity 40-90%) within 60 mins as quickly as possible.
3Sample diluent is used immediately after being opened.
WARNINGS AND PRECAUTIONS
.The kit should be sealed and protected against moisture.
.All positive specimens shall be validated by other methodologies.
.All specimens shall be treated as potential pollutant.
.DO NOT use expired reagent.
.DO NOT interchange reagents among kits with different lot No..
.DO NOT reuse test cards and any disposable accessories.
.Misoperation, excessive or little sample can lead to result deviations.
.As with any assay employing mouse antibodies, the possibility exists for interference by human anti-mouse antibodies (HAMA) in the specimen. Specimens from patients who have received preparations of monoclonal antibodies for diagnosis or therapy may contain HAMA. Such specimens may cause false positive or false negative results.
.This test result is only for clinical reference, should not serve as the only basis for clinical diagnosis and treatment, the patients clinical management should be comprehensive consideration combined with its symptoms, medical history, other laboratory examination, treatment response, epidemiology and other information.
.This reagent is only used for serum and plasma tests. It may not obtain accurate result when used for other samples such as saliva and urine and etc.
|Linearity||0.5ng/mL to 50ng/mL||relative deviation:-15% to +15%.|
|Linear correlation coefficient:(r)≥0.9900|
|Accuracy||The recovery rate shall be within 85% – 115%.|
|Specificity(None of the substances at the interferent tested interfered in the assay)||
1.Hansen JH,et al.HAMA Interference with Murine Monoclonal Antibody-Based Immunoassays[J].J of Clin Immunoassay,1993,16:294-299.
2.Levinson SS.The Nature of Heterophilic Antibodies and the Role in Immunoassay Interference[J].J of Clin Immunoassay,1992,15:108-114.
Key to symbols used:
|In Vitro Diagnostic Medical Device|
|Store at 2-30℃|
|Do Not Reuse|
|Consult Instructions For Use|
Xiamen Wiz Biotech CO.,LTD
Address:3-4 Floor,NO.16 Building,Bio-medical Workshop,2030 Wengjiao West Road,Haicang District,361026,Xiamen,China
Product detail pictures:
Related Product Guide:
Prostate cancer test you might need | P24 Test Strips
‘Screening Smarter’ for Prostate Cancer | P24 Test Strips
Our personnel are always inside the spirit of "continuous improvement and excellence", and together with the outstanding excellent goods, favorable price and good after-sales services, we try to gain every customer's trust for PriceList for Fob Tumor Marker Rapid Test Kit - Diagnostic Kit for Progesterone (fluorescence immunochromatographic assay) – Baysen, The product will supply to all over the world, such as: azerbaijan , Guatemala , Los Angeles , With a wide range, good quality, reasonable prices and stylish designs, our items are extensively used in this field and other industries. We welcome new and old customers from all walks of life to contact us for future business relationships and achieving mutual success! We welcome customers, business associations and friends from all parts of the world to contact us and seek cooperation for mutual benefits.
By Marian from Bahamas - 2016.12.02 14:11
This is a honest and trustworthy company, technology and equipment are very advanced and the prodduct is very adequate, there is no worry in the suppliment.
By Rae from Algeria - 2015.06.28 19:27